Installed my station yesterday (it has been sitting in the box since delivered last May, ID 6186 S/N HB-00004687, as of today running v103). While the rain reports were ~2x high during the storm that started yesterday (understandable with all the previous reports of similar performance with old firmware) it has not stopped raining according to the sensor. I have not seen any reports of this “feature” in the Forum. I assume my update to v103 took place overnight while it was raining which may have something to do with the new calibration routine being off for a baseline. The link below points to the WeatherUnderground daily data from yesterday.
Any suggestions before I take the batteries out of Sky and see if a reboot fixes this?