Looks like the system had enough data to apply a calibration on Saturday. How’s it look today, Ben?
It’s still like 3/4 hPa higher than the local official weather station… http://www.bom.gov.au/products/IDV60901/IDV60901.95874.shtml
(post withdrawn by author, will be automatically deleted in 24 hours unless flagged)
Hi Ben. We’ve discovered an issue with the OTA calibration process that looks like it’s affecting your Tempest along with several others. We’re working on a solution now…
Thanks. I always thought it was a software issue.